If you work at the bench, accurate pipetting is crucial—without accurate it your experiments would be non-reproducible, stock solutions inaccurate and assays would have such large errors that comparing them would be meaningless. But luckily, there’s no need to worry—your trusty, precision micropipettes take care of all that for you.
Or do they?
Precision instruments they may be, but the accuracy of micropipettes depends on you. You need to maintain your pipettes, practice good technique and have an understanding of how they work. Only once you have mastered those aspects can you claim to have anything like precision instruments at your disposal. Here’s how:
Know How Your Pipette Works
1. You Use an Air Displacement Pipette, Most Likely
An air displacement pipette works a bit like a syringe, except that there is an air-filled cushion between the piston and the sample. The air cushion prevents the piston from coming into contact with the solutions, which is good, but it also puts some limitations on the pipette.
The volume of the air cushion is affected by temperature and pressure, and volatile solvents can evaporate into it. Each of these affects pipetting accuracy. The barrel of air displacement pipettes is also vulnerable to contamination by the pipetted solution. If you are working with corrosives or bio-hazardous material, this can be a problem.
2. Don’t Use an Air Displacement Pipette (depending on your application!)
Most of the information in this article relates to air displacement pipettes, but in certain situations a positive displacement pipette may be a better option. Positive displacement pipettes also work like a syringe but, unlike air displacement pipettes, they don’t have an air cushion. This makes them more accurate for pipetting volatile solvents and more suitable for pipetting corrosives and bio-hazardous material. They are expensive because the barrel is replaced as part of the tip, but can be a good option in some cases. A cheaper alternative is to use and air displacement pipette with barrier tips, but these only address some of the problems.
However, you made need to have it services more often depending on your requirements for accuracy. The service should include re-calibration, greasing of the moving parts, and replacement of any worn out seals or other parts. It’s best to have this done by an experienced pipette doctor. If you work in university or large company, there may be pipette clinics you can use. In other cases, you can send your pipettes off to companies who will do it for you.
4. Check Your Pipettes Daily for Damage
Examine the nose of the barrel (where the tip is fitted) for any obvious damage. If there is a problem, have it serviced, because it is unlikely to be fit for the job.
5. Clean Your Pipettes Each Day Before Use
Wiping your pipette with some 70% ethanol should do it.
6. Store Your Pipette Vertically, Using a Pipette Holder
Storing your pipette this way prevents any liquids that have gotten into the pipette barrel from getting any further inside and corroding it.
7. Never Put Your Pipette on Its Side With Liquid in the Tip
The liquid might get into the pipette barrel and cause some serious corrosion damage.
8. Use Well-Fitting Tips
Poorly fitting tips allow air to escape when drawing up and dispensing liquid, leading to inaccurate results.
Hold the pipette vertically when drawing liquid in
Only immerse the tip slightly when drawing liquid in—otherwise you will coat the outside of the tip with liquid, which will be transferred along with the volume inside the pipette
When dispensing the liquid, hold the pipette vertically but keep the sidewall of the receiving vessel at 45 degrees. Pipette against the sidewall or into the liquid that’s already there.
10. Test Your Accuracy
Check the accuracy of your pipetting technique by dispensing 100 µL onto a fine balance. The mass of the droplet you make should be around 0.1 g. Now do the same thing 10 times and record the masses you obtain. If the variation is more than +/- 0.5% you need to either need to re-evaluate your pipetting technique or practice more!
Know the Tricks That Increase Your Accuracy
When you dispense liquid from your pipette, a coating of the liquid is left on the tip and makes the expelled volume slightly less than what is displayed on the pipette. Pre-wetting the tip before you pipette will help increase your accuracy. To pre-wet your tip, draw up the liquid into your pipette then dispense back into the original vessel. The coating is on the tip. Now, when you now draw up the liquid again and dispense it into the receiving vessel, none of it will be lost to wetting. This is only recommended for volumes greater than 10 µL.
12. Reverse Pipetting
Use this technique when pipetting viscous liquids or volatile solvents. Push the piston down to the purge position (the second stop), then draw the liquid up. There is too much liquid in the tip at this point but when the liquid is dispensed by pushing the piston to the aspirate position (the first stop), the extra liquid is left inside the tip. Using this method the tip is “automatically” pre-wetted but the extra liquid also helps when pipetting volatile solvents because some of the solvent will tend to evaporate into the air cushion.
13. Take the Ambient Temperature Into Account
The person or company that calibrated your pipette likely did so at room temperature. If you are working at a different temperature (e.g., in a cold room), then your pipette will not be dispensing the displayed volumes.
14. Take the Sample Temperature Into Account
In a recent Nature Methods publication, Millet and Barthlen observed a strange phenomenon. When repeatedly pipetting cold samples, the first dispensed volume is always larger than expected, but subsequent pipetting with the same tip gave the correct volume. The same was true for hot samples, except that the first dispensed volume was smaller than expected. Their solution was simple—dispense the first volume back into the original vessel, then start pipetting.
15. Use a Sensible Pipette for the Volume You Want to Dispense
The accuracy of your pipette decreases as the dispensed volume approaches the minimum the pipette can handle. For example, if you are dispensing 15 µL, then a 1 mL pipette would be terrible, a 200 µL pipette not so good, and 20 µL pipette ideal.
16. Use the Largest Volume Possible
Large volumes are easier to pipette accurately than small. Say you are performing an assay where you have to accurately pipette 5 µL. Pipetting that small amount accurately is not easy and will likely contribute greatly to the statistical error in your results. On the other hand, you could dilute the stock solution 10 times and pipette 50 µL of the solution. You could easily and accurately pipette this amount, which would yield much tighter error bars.
17. And Finally, Here’s How to Get Maximum Accuracy
An analytical balance is more accurate than any pipette. For maximum accuracy, use your pipette to dispense the volume you need but do it into a tared container on a balance. Then, calculate the actual pipetted volume from the mass using the formula volume=mass/density. Of course, this only works for solutions of known density—but for aqueous solutions that have a density of 1, this is not a problem.
If you work at the bench, accurate pipetting is crucial—without accurate it your experiments would be non-reproducible, stock solutions inaccurate and assays would have such large errors that comparing them would be meaningless. But luckily, there’s no need to worry—your trusty, precision micropipettes take care of all that for you. Or do they? Precision instruments […]
It’s great to have you in the Bitesize Bio family! We’ve sent you an email to confirm your registration. Please click on the link in the email or paste it into your browser to finalize your registration.
For more information on how to use Bitesize Bio, take a look at the following image (click it, for a larger version)
An error occured while registering you, please reload the page and try again